HAMEG INSTRUMENTS HM | Module: function generator; 7-Segment,5- digit LED; Band: ≤10MHz – This product is available in Transfer Multisort. F requency Range 50 mHz 10 MHz,. Output Voltage up to 10 Vpp (into 50 Ω). ☑ Waveforms: Sine Wave, Triangle, Square Wave, Pulse, DC. ☑ D istortion Factor. Funktionsgenerator HM Equipment HAMEG GmbH, Frankfurt, build ?, 4 pictures, 4 schematics, Germany, tubes, semiconductors, Service- or Lab.
|Published (Last):||3 May 2008|
|PDF File Size:||10.87 Mb|
|ePub File Size:||7.57 Mb|
|Price:||Free* [*Free Regsitration Required]|
Corres- ponding to Safety Class 1 regulations three-conductor AC power cable. Adjust 0, until 1.
The display “” cannot and “” should not be exceeded. The signal amplitude then extends over about 5 div. These free programs can be 0830 on this page: After unpacking the module, check for any mechanical damage or loose parts inside. This file is downloadable free of charge: Frequencies are read out on a 4 digit LED display.
HM8030 Function Generator Module
Instead, contact your nearest service center! If exact square-shaped signals are required, care should be taken that only 50 ohms coaxial cables e. The product does not contain any of the restricted substances in concentrations and applications banned by the Directive, and for components, the product is capable of being worked on at the higher temperatures required by lead—free soldering.
These val- ues are obtained when the generator output is terminated with 50 hamg. When later replacing the case, care should be taken to ensure that it properly fits under the edges of the front and rear panels. F – Distortion adjustment Setting: Signal waveform at the trigger output Setting: Set the trigger edge con- trol slope of the oscilloscope to negative edge trigger- ing.
Thank You for Choosing Test Equipment Depot – Hameg No Longer Available.
The TP 1 test point is located next to the amplitude control on the upper PC board. When removing or replacing the metal case, the instru- ment must be completely disconnected from the mains supply.
By clicking the accept button below, you agree to the following terms. The selected frequency appears on the 4-digit display 0. HAMEG is not liable for consequential dam- ages. When the knob is in CAL. With the maximum amplitude of DC offset If the offset knob is not in its calibrated position, a DC voltage can be superimposed on the output signal. Intentional interruption is prohibited. Good luck to the repair!
HAMEG HM Function Generator 10mhz | eBay
If test circuits having a 50 ohms input impedance are con- nected, this termination is not required. Please supply name and department, address and telephone number to ensure rapid service.
Additional quality fea- tures include the relatively low distortion factor of the generated signals and constant amplitude flatness throughout the entire frequency range of the instrument.
A small circle o is now revealed on the red power button in the front centre of the mainframe. Accord- ingly, the overdrive indicators are light up in case of posi- tive or negative overdriving. This permits the user to change the sweep width. The instrument should be returned in its original packaging for maximum protection. Any interruption of the protective conductor inside or outside hamrg instrument hzmeg disconnection of the protective earth terminal is likely to make the instrument dangerous.
HM function generator module,5MHz. H 68, D mm Weight: Adjust timebase, so that the positive pulse period of the output signal has a width of 803 10 divisions. If any measurement or calibration procedures are unavoidable on the opened-up instrument, these must only be carried out by qualified personnel acquainted with the danger involved. The frequency changes by a factor of Set oscilloscope to DC coupling. Any zeroes preceding the decimal point are dropped.
Hamfg the result of step 1 and if necessary repeat the adjustment. Before opening the set, you should carefully read the Safety and Warranty information on page M2 of these operating instructions.
Subject to charge without notice M3 – 2 Operation Function selection The type of output signal is selected with the function selection switch A total number of 3 different wave- forms – sine, square and triangle – are available. Setting range from x0. Connect the distortion meter to TP1. Use a 50 ohms through-termination.
Alternatively adjust 0 and 0, until the minimum value is obtained. If these precautions are not observed, overshoot may occur, especially when high frequencies are selected.
The Hz and kHz range indicators are integrated into the dis- play panel. It is TTL compat- ible and has a duty-factor of approx. Frequency variation by FM input Setting: No other warranty is expressed or implied.
Pull out the previously used module by its handle and slide in the new module until the end position is reached. Should there be any trans- portation damage, inform the supplier immediately and do not put the module into operation. If necessary, remove dust from inside the module using a small brush.
FM input If a positive DC voltage is applied to the FM inputthe generator frequency increases and is accordingly dis- played.